|Date Added:||21 March 2005|
|File Size:||57.58 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Configuring Oscilloscope timebase from labview – Stack Overflow
Frequently Asked Questions Q1: Sign up using Facebook. These are 8-wire connectors slightly larger than RJ modular phone jacks. Back to Top 3. Testing Now we need to verify that we configured everything correctly and that our cable is working okay.
Post as a guest Name. Yes, but you must use a hub in this case.
Tektronix TBS Digital Storage Oscilloscope – Instrument Driver – National Instruments
On the computer, navigate to http: Rate this gektronix Select a Rating 1 – Poor 2 3 4 5 – Excellent. Now we need to verify that we configured everything correctly and that our cable is working okay.
LabVIEW Add-On for Tektronix Open Windows Oscilloscopes Version 1.2
In this dialog, we want to turn off Obtain an IP address automaticallyand switch it to Use the following IP address. Be sure that the IP address and Subnet mask match the numbers shown above. This should bring up a page that looks something like oscilloscppe. A cross-over cable maps the 1,2 pair on each end to 3,6 on the tekgronix end. This simple diagram should return a typical response from the instrument.
This should bring up the window below. Can I connect three or more devices together?
Configuring a Tektronix Oscilloscope to Communicate with a Crossover Ethernet Cable
Sign up using Email and Password. Everything else is the same. Email Required, but never shown.
For those who may have this question, I do have found a way around. This document describes how to set up a computer and Ethernet-based instrument labviee communicate over a simple Ethernet cross-over cable. Ashutosh 1 4 In the NI driver for scope to set the timebase “Configure Timebase. Press OK Accept to apply the changes.
First, you need to get to the network configuration control panel.